100% tested for flow, leaks and connector retention
Gas specific rough-in accepts either: Puritan-Bennett, Ohmeda, Chemetron, DISS, Oxequip/Medstar
Click on the photos below to enlarge
Hose Barb Outlets: The gas inlet tube of the outlet body shall be a hose barb fitting to permit connection of either 1/4" or 5/16" or 3/8" O.D. flexible hose with a crimp type ferrule.
DISS Outlets: Gas specific DISS connection brazed to the back body of the gas outlet which will allow a gas specific hose to be attached to the connection.
The Medical Gas Ambulance outlet shall be an Amico Alert-1 series.
Medical Gas Ambulance outlets shall be gas specific and permanently labelled for the services indicated.
All outlets meet applicable requirements of "Federal Specification for Ambulances" KKK-A-1822C. Amico ambulance outlets comply with NFPA-99.
For positive pressure gas services, the outlet shall be equipped with a primary and secondary check valve and the secondary check valve shall be rated at a maximum of 200 psi (1,379 kpa) in the event the latch valve assembly is removed for maintenance.
Outlet bodies shall be gas specific by indexing each gas service to a gas specific dual pin indexing arrangement on the respective identification module.
A large color coded front plate shall be used for ease of gas identification and aesthetic appeal.
The latch-valve assembly shall be Ohmeda Diamond Quick Disconnect, Chemetron Quick Connect, Puritan-Bennett Quick Disconnect, DISS compatible or Oxequip/Medstar compatible and accept only corresponding type gas specific adapters.
All outlets shall be cleaned and degreased for medical gas service, factory assembled and tested.