Customizable to meet your facility’s specific needs.
Expandable to meet your future requirements.
Powder coated access panels and laminate colors match your room décor
Services manifolded to a single point for ease of installation
Recommended for all acuity levels including:
Flexibility Meets Your Needs
Accessory rails can be used for IT, monitor, cord and equipment management to meet current needs and provide seamless upgrades for future technology
Easy to Clean
Rounded top and bottom covers are easy to clean and encourage better infection control
Injection molded end cap increases durability and makes it easier to clean
Upgradeable for Future Needs
Integrated raceway system allows for future upgrades
Optional rough in back plate to reduce installation time
Optional chase for surface mounted rough in
Materials and Construction
Four accessory rails which are integrated into the aluminum extrusion assembly. The accessory rails are clear etched anodized.
Laminated fascia is composed of aluminum strips with high pressure laminate (HPL), color and pattern selection by customer
Top, middle and bottom cover panels are made of power coated extruded aluminum. End caps are made of injection molded ABS Fire Retardant plastic.
Optional service chase is made of 6063-T5 extruded aluminum with passage space for conduit and medical gas piping. High Pressure Laminate MDF inserts shall finish the exposed front, and shall match the laminate color selected.
Medical gas distribution shall be supplied by pipe drops via the service chase or single point termination zone as indicated on the drawing. All Medical Gas Outlets and piping shall be brazed and tested in accordance with NFPA 99c.
Wire for standard and critical branch power circuits shall be #10 or #12 (as specified) type THHN stranded copper wire, 600 volt, with heat resistant thermoplastic insulation.
All ground conductors shall be installed in conduit. Each power receptacle shall have a ground conductor connected to a grounding screw. A grounding bus will be installed in each device barrier box at rough in locations, and shall be installed as to ensure grounding for the complete power system.
Electrical receptacles shall be Hubbell Hospital Grade 15 or 20 amp, 120 or 277 volt, UL listed and marked Hospital Grade. Simplex, duplex, and (optional) safety receptacles shall be NEMA style 5-15R or 5-20R. Optional locking receptacles shall be simplex type, 20 amp, 120 or 277 volt, color black, and be HUBBELL lock 23000 HG style.
Low voltage data provisions shall be connected to device junction box via conduit or raceway. Amico Corporation shall include pull cord extending from junction box to service provision.
Switching shall be HUBBELL Industrial Grade 120 or 277 volt, 15 or 20 amps. Switch type options include SPST, 3-Way or Momentary. Low Voltage Switching will be 0-12 volts, 15 amps unless otherwise noted. Amico Corporation shall furnish, preinstall and wire all switches.